This is because both programs are using the same data acquisition board, and running the MainProtocol in both programs would cause both programs to access the same board at the same time, and would be complete chaos. Otherwise contact National Instruments support. There are two wells per chamber, and each well contains one slice from a different animal. You can change the delay time 5 sec or 0. This is true for entering a solution change by keyboard input AFTER the first sweep following the solution change time as long as fewer than sweeps have occurred in between because WinLTP remembers the time of only the last sweeps. The first protocol file FastProtocol1.
|Date Added:||13 July 2007|
|File Size:||24.80 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Log in or register to post comments. These enabling check boxes work the same way as those for the Repeat P0 Sweeps. Also note that For LTD stimulation with averaging, the current and total number of sweeps evoked is now written on the RunLine. Therefore, on startup, the different WinLTP acquisition programs will be automatically placed in different screen locations. Because T0 sweep has S0 pulses and T1sweep has 1 S1 pulse, S0 pulses appear where a single S0 pulse would normally appear, and the S1 pulse frequency is not disrupted.
The View sweep magenta trace is then loaded not exactly over the Current sweep blue trace right arrow and the ADsweep filename is digidzta into the Current Viewed Sweep field upper left dkgidata.
Axon™ pCLAMP® and DigiData™: Operating system compatibility
The Which Stimulus Acquisition Sweeps dialog box determines which combination of stimulation acquisition sweeps appear in the MainPg area. P0, First sweep after AddedEvent Stimulus Artifact Blanking cannot be performed on Spontaneous Sweeps. Then use the menu command: Input must be in the form of AnalysisTime and message, “mm: Slow0 and Slow1 Perfusion can control two valve controllers, and the Fast0 Perfusion controls the Stepper which controls whether solution from Tube 0 blue or Tube 1 brown is applied to the cell.
Adding the four Analysis Graphs on the MainPg with the eight Analysis Graphs on the Analysis Page gives a total of twelve Analysis Graphs that can be viewed during online acquisition and reanalysis. And you can further refine the X and Y 132a values on the graphs by changing the associated field values. Repeat ason stimulation, acquisition and analysis b. To change the acquisition sweeps that appear on the MainPg area use the View menu command Fig.
Choosing standard single-line Fast Perfusion in the automated perfusion control radiobutton group.
Make sure to remove the connections to the Analog Outputs before calibrating. B LTD stimulation with one extra Delay period. Remember, these will be the values in the program when the program is restarted after a crash. Two sweep stimulations are Pulse Sweeps P0 and P1, and these are usually used for single pulse stimulation, can be repeated at set time intervals, and ddigidata sweep data can be signal averaged. Note that A1 contains a FlushTime edit in seconds to change the duration of pre-flushing the new perfusion line.
This file can then be used for off-line analysis of spontaneous events with other programs Fig. This is what happens with LTP stimulus train induction without writing any scripts.
For faster solution changes typically down to a millisecond in the new solution, a mechanical stepper device to move between solution pipes is required Digiidata The next field to the right is the period of the sweep in seconds. Whether you need to use the pre-flush system that requires twice as many valves as the standard system depends on your perfusion requirements.
After the MainProtocol was run, the A Analysis of a train using baseline of the first pulse and response of the last pulse. The normal value is A, automatic. You might as well download and use the latest Version 9.
Axon Instruments DigiData 1322A 16-bit Data Acquisition System
Digidatw Electrode Series Resistance Rs 9. An example of no time between sweeps. This same idea can be used when using Sequential Scripts in the Protocol Builder.
This is particularly useful when wanting to view the sweep from an earlier part of the experiment, or a sweep that generated an outlier data point. The researcher can leave at this time. The Sweep Acquisition graph shows a fEPSP wxon by S0 stimulation 10 ms after the start of the sweep, averaged from 4 sweeps with a red line showing the slope.